0.02 to 90L/h and 4500:1 wide range measurement (H type)
Keep the temperature change of the circulation flow passage in the device within ±0.1℃ (variation per 1 minutes)
to provide stable measurement.
Pipe arrangement which minimizes air bubbles that enter the system reduced the factor error causes.
Resolution has been improved by using an encoder with 1200P/R (Ten times of number of pulses to the existing model.)
|Model name||Product name||Remarks|
|1||Fuel measurement system|
|2||Fuel supply station|
|3||FM-2500A||Digital flow meter|
Example of measurement data
Stability of temperature control by constant water circulation
● The stability of temperature is improved by the use of constant temperature water that is prepared in advance,
not using the control temperature by both hot water and cold water.
Accuracy of integrated fuel flow when temperature is stabilized
● Accuracy of integrated fuel flow measured by several detectors same as that of fuel flow measured by a single detector is
achieved by stabilizing the temperature of fuel circulation.
*Data when the FP-213S (1200 P/R) used.