Analog signal is outputted as follows in the SS measurement mode (measurement mode which displays max. value, min. value or average value).
1. When the SS measurement mode is selected.
(Press MENU button and select MEASURE > MEASURE MODE > SS in this order)
Analog signal is not outputted.
2. When the measurement starts by pressing the START button.
Analog signal regarding to the revolution which is updated in every 10 ms is outputted.
3. When the measurement stops by pressing the START button.
The analog signal when the measurement is ended is continuously outputted.
4. When the RESET button is pressed.
(The condition is returned to the state before the start of SS measurement mode)
The Analog signal is not outputted.