If the pulse number of the combined sensor for TM-2100 is not clear, when replacing the TM-2100 with the TM-3100, it is possible to setup the display value (r/min) same as the TM-2100 with the following methods.
1. DisplaySetup the [MEASURE > FACTOR] value of the TM-3100 same as [multiplier × index] of the TM-2100. (Setup the pulse number same as the default value of 60.)
2. Analog outputExample of the FULL value of the TM-3130 analog output When the frequency range of TM-2130 is 5 kHz: FULL rotation speed (r/min) of TM-3130=5000
The decimal point, input coupling and others can be obtained by the TM-2100 setup.
Caution; The above setup can be used when the display unit is r/min.